- Technical advance
- Open Access
Overlay of conventional angiographic and en-face OCT images enhances their interpretation
BMC Ophthalmology volume 5, Article number: 12 (2005)
Combining characteristic morphological and functional information in one image increases pathophysiologic understanding as well as diagnostic accuracy in most clinical settings. En-face optical coherence tomography (OCT) provides a high resolution, transversal OCT image of the macular area combined with a confocal image of the same area (OCT C-scans). Creating an overlay image of a conventional angiographic image onto an OCT image, using the confocal part to facilitate transformation, combines structural and functional information of the retinal area of interest. This paper describes the construction of such overlay images and their aid in improving the interpretation of OCT C-scans.
In various patients, en-face OCT C-scans (made with a prototype OCT-Ophthalmoscope (OTI, Canada) in use at the Department of Ophthalmology (Academic Medical Centre, Amsterdam, The Netherlands)) and conventional fluorescein angiography (FA) were performed. ImagePro, with a custom made plug-in, was used to make an overlay-image. The confocal part of the OCT C-scan was used to spatially transform the FA image onto the OCT C-scan, using the vascular arcades as a reference. To facilitate visualization the transformed angiographic image and the OCT C-scan were combined in an RGB image.
The confocal part of the OCT C-scan could easily be fused with angiographic images. Overlay showed a direct correspondence between retinal thickening and FA leakage in Birdshot retinochoroiditis, localized the subretinal neovascular membrane and correlated anatomic and vascular leakage features in myopia, and showed the extent of retinal and pigment epithelial detachment in retinal angiomatous proliferation as FA leakage was subject to blocked fluorescence. The overlay mode provided additional insight not readily available in either mode alone.
Combining conventional angiographic images and en-face OCT C-scans assists in the interpretation of both imaging modalities. By combining the physiopathological information in the angiograms with the structural information in the OCT scan, zones of leakage can be correlated to structural changes in the retina or pigment epithelium. This strategy could be used in the evaluation and monitoring of patients with complex central macular pathology.
The combined use of morphological and functional information has been shown to improve diagnostic accuracy in many clinical disciplines [1–3]. In the management of patients with malignancies, coronary heart disease and diseases of the brain, fusion of different image modalities, such as CT, MRI, PET and various other nuclear diagnostic tests, is widely used [3–7]. Retinal diseases are also very complex, and often more than one diagnostic or imaging technique is used to assist in making a diagnosis. Cunha-Vaz and co-workers [8, 9] have shown the value of multimodal mapping systems for the macula, incorporating retinal imaging techniques such as confocal scanning laser ophthalmoscopy (SLO), the retinal leakage analyzer, the retinal thickness analyzer, automated perimetry and SLO angiography. However, inherent difficulties exist in the ability to correlate these various modalities, as these techniques lack common, precise reference points.
Recently, a new modality in ocular imaging was introduced combining high depth resolution OCT with high transversal resolution confocal ophthalmoscopy [10–13]. The prototype OCT-Ophthalmoscope simultaneously produces en-face OCT scans and pixel-to-pixel corresponding confocal images, so called OCT C-scans. The OCT C-scans are oriented in a transversal plane, perpendicular to the optical axis of the eye. A stack of such OCT C-scans allows for a reconstruction of the retina in three dimensions. Although this transversal plane is the more familiar plane when viewing the retina, compared to the conventional, longitudinal OCT images [14–16], single en-face OCT scans may look confusing to the uninitiated, due to their high depth resolution (about 10 micron) and their transversal orientation through the retina [12, 17].
The confocal channel creates a fundoscopic image with a high transversal resolution but a low depth resolution (~300 μm). There is seemingly little change in the confocal image when scanning through the retina along the Z-axis (depth). Therefore the confocal part of the OCT C-scan provides a high quality fundus image throughout the whole OCT stack in depth. As both the confocal and OCT image are acquired simultaneously with the same light source and at the same scanning rate, there is pixel-to-pixel correspondence between the OCT and the confocal image of the OCT C-scan . The confocal image is used for general orientation and localization, whereas the OCT image shows detailed morphology, and subsequent pathological changes within the retina .
The high quality confocal image makes it a reliable reference image to overlay the OCT C-scans onto images generated with other diagnostic techniques acquired in the transversal plane. For example, with appropriate software to implement an affine transformation, conventional angiographic images can be spatially transformed and then superimposed over the confocal image. Given the pixel-to-pixel correspondence between the confocal and OCT channel, the converted angiographic image can be directly superimposed over the OCT image as well. The fusion of the more familiar angiographic information with the highly detailed morphological information from the OCT C-scan could improve our knowledge of retinal diseases as visualized by en-face OCT. In this paper, we describe this novel procedure, and assess its ability to enhance our understanding of both fluorescein angiography and OCT C-scans through the combination of structural and functional information.
At the Department of Ophthalmology of the Academic Medical Centre (University of Amsterdam, Amsterdam, The Netherlands) a prototype OCT-Ophthalmoscope (Ophthalmic Technologies Inc., Toronto, Canada) was used to evaluate patients with various retinal pathologies. Patients undergoing routine fluorescein angiography (FA), using Imagenet (Topcon, USA), were scanned with the OCT-Ophthalmoscope in the area of interest on the same day. To demonstrate the possibilities of this new technique, we selected some illustrative, though not necessarily common, pathological cases.
The configuration of this OCT prototype has been described previously [10, 13]. In short, the system uses a super luminescent diode, with a central wavelength of 820 nm (20 nm bandwidth). The light beam is split, directing one part to the patient's eye (sample arm) and the other part to a reference arm (mirror). The returning light beams from both the patient's eye and the reference arm are collected through an interferometer to produce the OCT signal . A fraction of the light returning from the patients' eye is also directed towards another detector to produce a confocal signal . The OCT-Ophthalmoscope produces a transversal OCT C-scan (size: 1042 × 512 pixels), currently at 2 frames/second, in the X-Y-plane at a fixed Z-coordinate, in which the OCT and confocal images are in pixel-to-pixel correspondence with each other .
ImagePro is an image-processing program working under Windows©. By using a custom made plug-in for this program, it is possible to spatially transform images, and to superimpose them over reference images. For the purpose of this study, the confocal part (size: 512 × 512 pixels) of the OCT C-scan was used as a reference image. Since every transformation inherently leads to loss of anatomical resolution in the transformed image, we chose to leave the OCT C-scan intact. The OCT C-scan covers a smaller area than the FA. Detailed information in the OCT scan would be lost due to a necessary volume transformation if the OCT C-scan would be transformed onto the FA. One or two OCT C-scans were chosen from the stack, depending on the depth at which the pathologies were best visible. The appropriate angiographic Imagenet frame was transformed onto, and superimposed over the confocal image. Alignment was then achieved by using evident landmarks as reference points. Inherent to the scanning procedure of the OCT-Ophthalmoscope, using a curved fan ray equivalent to the curvature of the eye, there is some distortion at the borders of both parts of OCT C-scan relative to the centre of the images. We found that it was best to choose reference points centrally within the vascular arcades. Four to six pairs of reference points were manually selected. This was done by zooming in on the area in which a landmark was localised. Landmarks were chosen at vascular crossings or vessel bifurcations that were clearly visible in both image modalities. These reference points were used to compute a linear transformation by the least-square method, compensating for translation, rotation, scaling, and shearing (full affine). The spatially transformed angiographic image was doubled and then superimposed over both the confocal image and the OCT scan. To further facilitate visualization, the images were combined into the different layers of an RGB image: the green channel of the RGB image was used for the spatially transformed angiographic image, and the other channels were used for both parts of the OCT C-scan. Images presented in this paper show on the left-hand side the confocal image, and on the right-hand side the OCT image with or without angiographic overlay. Figure 1 shows the above described procedure for making an overlay image with the help of OCT C-scan and FA images of a normal fundus. The whole procedure (including acquisition and selection of the appropriate image frames) took no more than 15 minutes per patient.
Example A: Birdshot retinochoroiditis (figure 2)
The OCT C-scan shows marked retinal swelling temporal to the fovea (figure 2a), with small cystoid changes. On the late FA image (figure 2b) leakage is seen along the vascular arcades and in the fovea. The spatially transformed FA image (figure 2c) is superimposed over the OCT C-scan (figure 2d). This overlay image shows that the temporal area of retinal thickening on the OCT exactly matches the temporal, parafoveal area of leakage on the angiographic image. Reviewing the overlay image more closely, smaller areas of extensive leakage at the nasal side of the fovea and along the vascular arcades on the angiographic image (figure 2c, asterisks) correspond to the somewhat irregularly shaped retinal contour on the OCT C-scan (figure 2a, asterisks). Generalized retinal thickening along the arcades corresponds to extensive intraretinal vascular leakage on the angiogram. There is correspondence between a physiologic parameter (leakage) and morphological changes (retinal thickening and cystic changes).
Example B: high myopia with secondary neovascularization (figure 3)
Figure 3a shows an OCT C-scan at the level of the retinal pigment epithelium (RPE) in a highly myopic patient with a solitary neovascular membrane, located just outside the foveolar area. An irregularly shaped, hyperreflective circular line is seen, corresponding to a pigment epithelial detachment (PED). The content of this circular area is dense, suggesting that it contains a highly reflective tissue such as a neovascular membrane rather than serous fluid as might be seen in central serous retinopathy . In addition, on the OCT C-scan, the distinct sharp border generated by the PED is surrounded by a dark halo. On the OCT B-scan, the inner band of the double highly reflective layer at the level of the RPE is interrupted just before the edges of the PED (figure 3b), corresponding to the dark halo on the OCT C-scan. The dark halo on the OCT C-scan and the interruption of this inner highly reflective layer in the OCT B-scan is most likely caused by the changed orientation of the photoreceptor outer segments. This changes the reflectivity properties of the photoreceptor layer and therefore the reflectivity profile in the OCT image. The FA image (figure 3c) shows very prominent leakage in the centre of the lesion, surrounded by a first halo of hypofluorescence, and a second halo of hyperfluorescence. Good spatial correspondence is seen in the overlay (figure 3e). The bright spot of prominent leakage on the FA is located well within the confinement of the PED as seen in the OCT C-scan. The hypofluorescent halo on the FA corresponds to the hyperreflective circle on OCT delineating the RPE surrounding the neovascular membrane. The outer hyperfluorescent zone on the FA corresponds to the hyporeflective halo on the OCT. This hyperfluorescent zone on the FA shows the characteristics of a window defect. The corresponding hyporeflective halo in the OCT C-scan surrounding the membrane can be a result of this loss of pigmentation, but is more likely caused by changed reflectivity of the photoreceptor layer due to its changed orientation, as supported by figure 3b. Both OCT B- and C-scan also clearly show that the membrane is located below the RPE, which makes it ineligible for current surgical techniques.
Example C: Retinal Angiomatous Proliferation (RAP, figure 4)
This patient has a stage II RAP based on the FA image (figure 4b) . Reviewing the FA image, a large area of hyperfluorescence is seen surrounding a bright, fluorescent central spot, presumably the RAP. The first OCT C-scan (figure 4a, right) is taken at the level of the retinal vessels. It shows a large serous neurosensory detachment and a large, very irregularly shaped PED. The overlay shows a discrepancy between the hyperfluorescence in the FA and the structural information in the OCT image (figure 4d). The hyperfluorescence seems mostly confined to a part of the PED seen on the OCT. The bright spot within the area of diffuse hyperfluorescence can be correlated with a highly reflective area in a second OCT C-scan taken above the large PED (figure 4e–h). This appeared to correspond to the top of the PED, but more likely relates to a highly reflective proliferation located between the neurosensory retina and the RPE (not shown).
The OCT-Ophthalmoscope provides clinicians with a novel imaging modality of the retina. The transversally oriented OCT C-scans allow software-assisted overlay with more traditional imaging techniques, such as FA or indocyanine green (ICG) angiography, that are also displayed in a transversal plane. The confocal channel in the OCT-Ophthalmoscope provides a fundoscopic image with a high transversal resolution which has a pixel-to-pixel correspondence with the OCT channel that possesses a very high depth resolution . Therefore, the confocal image can easily be used as a reference image. Images produced by other diagnostic techniques can be mathematically transformed to fit snugly over the confocal image of the OCT C-scan, and thereby be directly superimposed over the OCT image.
Our initial goal in making overlay images, was to enhance the interpretation of individual OCT C-scans. These are initially difficult to interpret. The high transversal resolution makes recognition of retinal landmarks difficult in the OCT C-scan images. Classically, pathologic processes are studied in longitudinal sections, so that their transversal appearance is rarely appreciated. By using the overlay technique, it is possible to correlate pathologic features imaged with familiar transversal imaging techniques, such FA, with the morphological information provided on OCT C-scans. We also found that the overlay technique provided additional insight not readily available with either modality alone.
In our example of Birdshot retinochoroiditis (figure 2) the OCT C-scan showed an irregularly shaped retinal contour in the parafoveolar area, and along the arcades. On the FA image diffuse parafoveolar oedema was present as well as diffuse leakage along the major retinal vessels. Leakage correlated well in the overlay to the irregularly shaped retinal contour on the OCT. Thus, the patchy retinal thickening seen on OCT was caused by diffuse oedema, which if needed could be followed prospectively, without the need of the more invasive angiographic technique. In figure 3, OCT and FA concurred in the location of the neovascular membrane. Further, the OCT also showed that the neovascular membrane was situated below the RPE, making this membrane ineligible for current surgical procedures , while the retina above was intact and not involved in the process. Combining findings on the longitudinal and transversal OCT allowed one to better comprehend the FA manifestations observed at the edge of the neovascular lesion. Hypofluorescence at the edge of the PED is due to superposition of pigmented RPE cells, while the more peripheral hyperfluorescence is caused by a window defect. The OCT and FA images in the RAP example (figure 4) revealed the limitations of FA in defining the extent of involvement, in particular of the RPE [21, 22]. FA leakage was present in only a portion of the PED, probably due to the presence of blocked fluorescence, although the cause of this blocked fluorescence could not be ascertained with the OCT.
Conventional OCT is a valuable tool in making a diagnosis and in monitoring the progress of macular diseases. The commercially available StratusOCT scans in a longitudinal orientation, providing high resolution, morphologic, cross-sectional OCT scans of the retina, but it does not provide a precise localization of the scanned lines within the macular region, because there is always a time delay between acquisition of the line and storage of the line with its associated fundus image. A retinal thickness map is based on the interpolation of the thickness measurements from six radial OCT scan lines. While these maps are useful to monitor patients, they cannot be combined with other transversal diagnostic techniques because both the map and the scan lines which it is made of, lack good reference points. For this and other reasons, conventional OCT has not yet been incorporated in multimodal mapping systems such as the one developed by Cunha-Vaz and co-workers .
Rosen and co-workers  were able to produce simultaneous ICG angiography and en-face OCT in an adjusted, experimental OCT-Ophthalmoscope. Here pixel-to-pixel correspondence was seen in both channels, which ultimately may be a better set-up than the technique proposed here. However, this combination will need further optimization before it becomes clinically applicable. At this moment, our examples suggest that multimodal mapping is possible between the OCT-Ophthalmoscope and existing imaging modalities, without various imaging modalities being incorporated into the system. Combining several imaging techniques provides complementary information which helps in interpreting and understanding the findings of each modality. Additional modalities which could possibly be fused with transversal OCT include microperimetry and (multifocal) electroretinography. These would provide functional correlations to structural alterations in the retina.
Combining conventional angiographic images and en-face OCT C-scans assists in the interpretation of both imaging modalities. By combining the physiopathological information in the angiograms with the structural information in the OCT scan, zones of leakage can be correlated to structural changes in the retina or RPE. Fusion of different imaging modalities can contribute to our understanding of complex retinal or choroidal diseases and it can optimize our patient management.
Forster GJ, Laumann C, Nickel O, Kann P, Rieker O, Bartenstein P: SPET/CT image co-registration in the abdomen with a simple and cost-effective tool. Eur J Nucl Med Mol Imaging. 2003, 30: 32-39. 10.1007/s00259-002-1013-0.
Ketai L, Hartshorne M: Potential uses of computed tomography-SPECT and computed tomography-coincidence fusion images of the chest. Clin Nucl Med. 2001, 26: 433-441. 10.1097/00003072-200105000-00011.
Pietrzyk U, Herholz K, Schuster A, von Stockhausen HM, Lucht H, Heiss WD: Clinical applications of registration and fusion of multimodality brain images from PET, SPECT, CT, and MRI. Eur J Radiol. 1996, 21: 174-182. 10.1016/0720-048X(95)00713-Z.
Eubank WB, Mankoff DA, Schmiedl UP, Winter TCIII, Fisher ER, Olshen AB, Graham MM, Eary JF: Imaging of oncologic patients: benefit of combined CT and FDG PET in the diagnosis of malignancy. AJR Am J Roentgenol. 1998, 171: 1103-1110.
Faber TL, Santana CA, Garcia EV, Candell-Riera J, Folks RD, Peifer JW, Hopper A, Aguade S, Angel J, Klein JL: Three-dimensional fusion of coronary arteries with myocardial perfusion distributions: clinical validation. J Nucl Med. 2004, 45: 745-753.
Krempien RC, Daeuber S, Hensley FW, Wannenmacher M, Harms W: Image fusion of CT and MRI data enables improved target volume definition in 3D-brachytherapy treatment planning. Brachytherapy. 2003, 2: 164-171. 10.1016/S1538-4721(03)00133-8.
Schettino CJ, Kramer EL, Noz ME, Taneja S, Padmanabhan P, Lepor H: Impact of Fusion of Indium-111 Capromab Pendetide Volume Data Sets with Those from MRI or CT in Patients with Recurrent Prostate Cancer. AJR Am J Roentgenol. 2004, 183: 519-524.
Bernardes RC, Lobo CL, Cunha-Vaz JG: Multimodal macula mapping: a new approach to study diseases of the macula. Surv Ophthalmol. 2002, 47: 580-589. 10.1016/S0039-6257(02)00355-7.
Lobo CL, Bernardes RC, Santos FJ, Cunha-Vaz JG: Mapping retinal fluorescein leakage with confocal scanning laser fluorometry of the human vitreous. Arch Ophthalmol. 1999, 117: 631-637.
Podoleanu AG, Dobre GM, Cucu RC, Rosen RB, Garcia P, Nieto J, Will D, Gentile R, Muldoon T, Walsh J, Yannuzzi LA, Fisher Y, Orlock D, Weitz R, Rogers JA, Dunne S, Boxer A: Combined multiplanar optical coherence tomography and confocal scanning ophthalmoscopy. J Biomed Opt. 2004, 9: 86-93. 10.1117/1.1627778.
Podoleanu AG, Jackson DA: Combined optical coherence tomography and scanning laser ophthalmoscopy. Electronics Letters. 2002, 34: 1088-1090. 10.1049/el:19980793.
Rogers JA, Podoleanu AG, Dobre GM, Jackson DA, Fitzke FW: Topography and volume measurements of the optic nerve using en-face optical coherence tomography. Optics Express. 2001, 9: 533-545.
van Velthoven MEJ, Verbraak FD, Podoleanu AG, Rosen RB, de Smet MD: Clinical application of multiplanar en-face OCT in single or overlay mode with angiographic images. 2004, Monduzzi Editore International Proceedings Division, Medimond S.r.l., Proceedings of the 5th International Symposium on Ocular Pharmacology and Therapeutics - ISOPT: 11-14 March 2004;Monte Carlo (Monaco): 183-190.
Hee MR, Izatt JA, Swanson EA, Huang D, Schuman JS, Lin CP, Puliafito CA, Fujimoto JG: Optical coherence tomography of the human retina. Arch Ophthalmol. 1995, 113: 325-332.
Huang D, Swanson EA, Lin CP, Schuman JS, Stinson WG, Chang W, Hee MR, Flotte T, Gregory K, Puliafito CA: Optical coherence tomography. Science. 1991, 254: 1178-1181.
Drexler W, Sattmann H, Hermann B, Ko TH, Stur M, Unterhuber A, Scholda C, Findl O, Wirtitsch M, Fujimoto JG, Fercher AF: Enhanced visualization of macular pathology with the use of ultrahigh-resolution optical coherence tomography. Arch Ophthalmol. 2003, 121: 695-706. 10.1001/archopht.121.5.695.
Podoleanu AG, Rogers JA, Jackson DA: 3D OCT Images from retina and skin. Optics Express. 2000, 7: 292-298.
van Velthoven MEJ, Verbraak FD, de Smet MD: The OCT-Ophthalmoscope, clinical aspects in central serous retinopathy. Ophthalmic Res. 2003, 35: s207-
Yannuzzi LA, Negrao S, Iida T, Carvalho C, Rodriguez-Coleman H, Slakter J, Freund KB, Sorenson J, Orlock D, Borodoker N: Retinal angiomatous proliferation in age-related macular degeneration. Retina. 2001, 21: 416-434. 10.1097/00006982-200110000-00003.
Zolf R, Glacet-Bernard A, Benhamou N, Mimoun G, Coscas G, Soubrane G: Imaging analysis with optical coherence tomography: relevance for submacular surgery in high myopia and in multifocal choroiditis. Retina. 2002, 22: 192-201. 10.1097/00006982-200204000-00010.
Yannuzzi LA, Slakter JS, Sorenson JA, Guyer DR, Orlock DA: Digital indocyanine green videoangiography and choroidal neovascularization. Retina. 1992, 12: 191-223.
Sato T, Iida T, Hagimura N, Kishi S: Correlation of optical coherence tomography with angiography in retinal pigment epithelial detachment associated with age-related macular degeneration. Retina. 2004, 24: 910-914. 10.1097/00006982-200412000-00011.
Dobre GM, Podoleanu AG, Rosen RB: Simultaneous optical coherence tomography-Indocyanine Green dye fluorescence imaging system for investigations of the eye's fundus. Optics Letters. 2005, 30: 58-60. 10.1364/OL.30.000058.
The pre-publication history for this paper can be accessed here:http://www.biomedcentral.com/1471-2415/5/12/prepub
The author(s) declare that they have no competing interests.
MvV carried out the acquisition of the OCT images, made the software-assisted overlay images and was responsible for writing this paper. KdV participated extensively in acquiring the overlay images and contributed to writing this paper, especially the methodology section. FV participated in the design of this study, contributed to the interpretation of the acquired overlay images and to writing the paper. CP participated in the design of this study and coordinated the acquisition of the overlay images. MdS participated in the design and coordination of this study, contributed in interpreting the images and in writing the paper. All authors read and approved the final manuscript.
About this article
Cite this article
van Velthoven, M.E., de Vos, K., Verbraak, F.D. et al. Overlay of conventional angiographic and en-face OCT images enhances their interpretation. BMC Ophthalmol 5, 12 (2005). https://doi.org/10.1186/1471-2415-5-12
- Optical Coherence Tomography
- Retinal Pigment Epithelium
- Fluorescein Angiography
- Optical Coherence Tomography Image
- Retinal Angiomatous Proliferation