- Research article
- Open Access
- Open Peer Review
Tono-Pen XL tonometry during application of a suction ring in rabbits
- Spyridon K Charisis†1Email author,
- Harilaos S Ginis†2,
- Georgios A Kounis†2 and
- Miltiadis K Tsilimbaris†1
© Charisis et al; licensee BioMed Central Ltd. 2008
- Received: 08 January 2008
- Accepted: 30 July 2008
- Published: 30 July 2008
The purpose of this study is to evaluate the use of Tono-Pen XL in measuring IOP during the application of a suction ring in rabbit eyes with manometrically controlled IOP.
Tono-Pen XL was calibrated against direct manometry in 10 rabbit eyes. A suction ring was then applied in 4 rabbit eyes and the IOP was determined manometrically during suction ring application at 350 mmHg vacuum pressure. Finally, in 6 catheterized rabbit eyes the IOP was measured with Tono-Pen XL during suction ring application at suction vacuum from 350 to 650 mmHg, while keeping actual IOP stable at 30 mmHg and 60 mmHg.
Linear regression analysis revealed that the Tono-pen XL was reliable for IOPs between 10 and 70 mmHg (R2 = 0.9855). Direct manometry during suction ring application showed no statistically significant variation of Tono-Pen XL readings when the incanulation manometry intraocular pressure changed from 30 mmHg to 60 mmHg and no statistically significant correlation between suction vacuum and IOP measurements.
Tono-Pen XL measurements are unreliable during the application of a suction ring on living rabbit eyes even when the actual IOP is forced to be within the validated range of Tono-Pen XL measurements. This inaccuracy is probably related to altered corneal and scleral geometry and stress.
- Central Corneal Thickness
- Differential Pressure Transducer
- Suction Vacuum
- Proparacaine Hydrochloride
- Suction Ring
Vacuum rings (also referred to as suction rings) are used to stabilize the globe during microsurgical operations such as refractive surgery. In order to achieve suction, the eye's shape needs to conform to the suction ring's shape and therefore as a result of their application-in the general case-the sclera and the cornea are deformed. This deformation results in increased intraocular pressure and increased corneal stress. In terms of clinical safety, an IOP increase of 43 mmHg during suction is considered safe . Moreover it has been suggested that the duration of this increase should be minimal. .
The potential complications related to the IOP increase during suction ring application raise the necessity for accurate IOP measurements during operations performed with the aid of a suction ring. Tono-Pen XL is a handheld, computerized, fast and easy to use tonometer, with a small (1,5 mm) transducer tip. It is equipped with a small head that makes it minimally affected by corneal surface abnormalities and corneal central thickness (CCT) [3, 4]. Furthermore, compared to other tonometers, Tono-Pen XL showed the least error in estimation of true pressure in New Zealand white rabbits . The Tono-Pen XL provides accurate IOP measurements in a range from 5 to 80 mmHg .
The purpose of this study is to test the accuracy of Tono-Pen XL in measuring IOP during the application of a suction ring in rabbit eyes. Additionally, a calibration of the Tono-Pen for the rabbit eye using direct manometry was obtained.
All experiments were performed at the Institute of Vision and Optics (University of Crete) using healthy albino rabbits that weighed between 3 and 4 Kg. Animals were treated in accordance to the ARVO Statement for the Use of Animals in Ophthalmic and Vision Research. Initial examination of all rabbits, including baseline measurements of the IOP, showed no pathological findings. Prior to the experiment, all rabbits were anaesthetized using subcutaneous injection of ketamine (40 mg/kg) and xylazine (7 mg/kg). Central corneal thickness (CCT) of each rabbit was measured before the experiment using a contact ultrasound pachymeter (Sonogage, Sonogage Cleveland, OH).
Tono-Pen XL calibration
Tono-Pen XL (Medtronic Solan, Jacksonville, Florida) was calibrated against direct manometry in 10 eyes of 10 healthy albino rabbits, under general anaesthesia. A sterile 22-gauge intravenous catheter needle, connected to a saline column and to a differential pressure transducer (0–5 psi, 1-ms response time; model 286–686; RS Components, Ltd., Taipei, Taiwan), was inserted in the anterior chamber through the paracentral cornea. The differential pressure transducer was built in with the electronic amplifier and a 12-bit A/D converter in a box communicating with a computer unit through a data interface (RS-232; RS Components, Ltd), and could perform an IOP measurement every 0,15 sec and record them for further processing . Two drops of proparacaine hydrochloride 0.5% were instilled onto the cornea once systemic anaesthesia was achieved, before catheter insertion. The height of the saline column was adjusted in order to get IOP readings ranging from 10 to 70 mmHg (in steps of 10) in the pressure transducer. At each pressure level the IOP was measured eight times using a Tono-Pen XL tonometer, at the 5% level, as indicated by the TonoPen. The lowest and highest values were disregarded and mean values of the remaining six were calculated.
IOP during suction ring application
Four healthy albino rabbits were enrolled in this experiment. We applied a suction ring, of the same style and configuration as that used in LASIK scaled down to fit rabbit's eye, on the animal's right eye. The suction ring was connected to the vacuum port of a LASIK microkeratome control unit (SCHWIND Eye-tech-Solutions & VIEW-POINT technology AG; Kleinostheim, Germany). A 22-gauge intravenous catheter needle was placed in the anterior chamber through the paracentral cornea, at the border of the suction ring's opening. The catheter was connected to the pressure transducer. Then the suction vacuum was turned on, at a vacuum pressure of 350 mmHg, for approximately 1 minute, while the differential pressure transducer was continually recording IOP.
Accuracy of Tono-Pen XL measurement during suction ring application
This experiment was performed on 6 eyes of 6 healthy albino rabbits. The suction ring connected to the adjustable suction pump was placed on rabbit's eye. Proparacaine hydrochloride 0.5% was instilled onto the cornea to achieve topical anaesthesia before the application of the suction ring. Once suction was achieved, the 22-gauge intravenous catheter needle, connected to the saline column and to the pressure transducer, was inserted in the anterior chamber of the eye through the paracentral cornea. The height of the saline column was adjusted in order to get IOP readings of either 30 or 60 mmHg from the pressure transducer. At each manometric IOP level the suction vacuum was adjusted from 350 mmHg to 650 mmHg in steps of 50 mmHg. At each step the IOP was measured eight times with the Tono-Pen XL. The lowest and highest values were disregarded and mean values of the remaining six were calculated. The rationale behind hydrostatic regulation of the pressure at two different levels was that the pressure after suction ring application is not constant but is rather characterized by an initial peak followed by a exponential decay. Hydrostatic regulation ensured that the actual IOP was constant during TONOPEN measurements. The procedure lasted approximately 10 minutes for each eye.
Real_IOP = 1.3472*TONO-PEN-7.0333 [mmHg]. The upper limit of Tono-Pen XL measurements was 90 mmHg.
In order to evaluate the possible contribution of central corneal thickness in Tono-pen XL's offset, we performed a regression analysis between normalized Tono-pen XL error and pachymetry (Fig. 1B). No such correlation could be documented (R2 = 0.03).
IOP during suction ring application
Tono-Pen XL during suction ring
As the anterior chamber was connected to the saline solution column, the differential pressure transducer recorded no IOP changes when increasing the suction vacuum from 350 mmHg to 650 mmHg. This finding is reasonable as fluid could either flow into or out of the eye through the catheter depending on the set height of the column and the forces on the ocular wall from the suction ring.
In eyes set at stable IOP 60 mmHg, the Tono-Pen XL readings are more consistent between 45 and 60 mmHg (Fig. 3D). No correlation between the normalized error (defined as difference between TONOPEN and transducer readings divided by transducer reading) and the pachymetry could be documented at 60mmHg (Fig. 3E; R2 = 0,0225). Again, the distribution of mean values of Tono-Pen XL readings seems to correlate with suction vacuum (Fig. 3F), however one-way Anova showed no statistically significant correlation (P = 0,095 > 0,05).
Finally, for each of the suction vacuum pressures, the independent sample T-test showed no statistically significant variation (P > 0,05) of Tono-Pen XL readings when the incanulation manometry intraocular pressure changed from 30 mmHg to 60 mmHg.
Even if complications related to IOP elevation during suction ring application in refractive surgery are rare, they raise the necessity for more knowledge about IOP changes during such procedures [2, 1, 8–10]. This knowledge could contribute to increase safety during refractive surgery, not only by influencing patient's inclusion criteria, but also by contributing to the design of new suction rings that work well without excessive increase of IOP.
The rabbit used as an animal model in this study has some known limitations. For example, compared to humans, the rabbit has a different structure to the cornea (no Bowman's) and outflow pathways and high ocular rigidity that could influence IOP behavior during suction ring application. Furthermore, the suction vacuum range used in this study is less than the vacuum settings used in LASIK.
A first step towards understanding IOP behavior during suction ring application would be accurate IOP monitoring. A handheld, accurate and minimally influenced by CCT electronic tonometer such as Tono-Pen XL, theoretically, could be a valid candidate for such IOP measurements [5, 6, 3]. We calibrated Tono-Pen XL against monometry in 10 animals and we confirmed that the device is accurate for IOP measurements in a range from 10 mmHg to 70 mmHg. These findings are in accordance with the work of others that have shown the accuracy of Tono-Pen XL in this range of measurements [5, 6].
The validated range of Tono-Pen XL measurements has an upper limit that is not big enough for IOP measurements during suction ring application. In the current experiment, when the suction ring vacuum was on, the manometricaly measured IOP was characterized by a peak that ranged from approximately 120 mmHg to more than 210 mmHg. IOP during suction ring application has already been measured in human donor eyes , in cats  and in porcine eyes  and was found superior to 90 mmHg. These findings are in accordance to our measurements. IOP levels of this magnitude by far exceed the range of measurements of Tono-Pen XL. In a recent study, Acosta et al, evaluated three different tonometers (including the TONOPEN) and concluded that TONOPEN was not accurate or reproducible in estimating IOP in rabbits over the range they tested. However, our findings suggest that an appropriate calibration curve can be obtained, as other investigators have also suggested.
Our initial measurements revealed that the pressure levels reached after suction ring application exceeded the measurement range of TONOPEN XL (90 mmHg). It is however possible that this is not the case with other vacuum ring designs. However, besides IOP rise, vacuum ring application results to increased corneal stress. It was hypothesized that even at lower pressure levels the corneal stress associated to the vacuum ring could affect the reliability of TONOPEN XL measurements.
In order to evaluate the tonometer's performance in the conditions that are created during suctioning, we developed an experimental setup that permitted us to set the actual IOP within the validated measurement range of Tono-Pen XL. After suction was achieved in catheterized eyes, we lowered and stabilized IOP to values that can be accurately measured by the Tono-Pen XL. We evaluated whether under these circumstances, the Tono-Pen XL can provide valid IOP measurement. Under these conditions Tono-Pen showed a big variation in measurements. Although there was the impression that Tono-Pen XL becomes more accurate when the real IOP increases, we were not able to document a statistically significant correlation between actual IOP and Tono-Pen XL measurements. This serious limitation of Tono-Pen XL, renders this device unreliable for IOP measurements during suction ring application not only because of its measurement range limits but also because of factors related to the special conditions developed during suctioning of the eye. It is possible that small differences in size and geometry of eyes may result in dramatic differences in the corneal stress distribution during suction ring application. Although a complete analysis of suction ring related corneal stress is a complicated problem, requiring accurate mechanical modelling, it is reasonable to hypothesize that variations in scleral and corneal geometry may result to variations in corneal stress and therefore to variations in the measured IOP with the Tono-Pen XL. It is also probable that the higher the IOP the less scleral and corneal geometry are altered by a suction ring application.
In conclusion, our results show that Tono-Pen XL measurements are unreliable during the application of a suction ring, probably because of altered corneal and scleral geometry and stress.
This paper is part of the 03ED375 research project, implemented within the framework of the "Reinforcement Programme of Human Research Manpower" (PENED) and co-financed by National and Community Funds (25% from the Greek Ministry of Development-General Secretariat of Research and Technology and 75% from E.U.-European Social Fund).
- Garhofer G, Resch H, Weigert G, Lung S, Simader C, Schmetterer L: Short-Term Increase of Intraocular Pressure Does Not Alter the Response of Retinal and Optic Nerve Head Blood Flow to Flicker Stimulation. IOVS. 2005, 46 (5): 1721-1725.Google Scholar
- Lester M, Tizte P, Mermoud A: Retinal nerve fiber layer thickness changes after an acute increase in intraocular pressure. J Cataract Refract Surg. 2005, 31: 590-594. 10.1016/j.jcrs.2004.06.085.View ArticleGoogle Scholar
- Bhan A, Browning A, Hamilton R, Dave D, Dua H: Effect of corneal thickness on intraocular pressure measurements with the pneumotonometer, Goldmann applanation tonometer, and Tono-Pen. Invest Ophthalmol Vis Sci. 2002, 43 (5): 1389-1392.PubMedGoogle Scholar
- Dohadwala A, Munger R, Damji K: Positive Correlation between Tono-pen Intraocular Pressure and Central corneal Thickness. Ophthalmology. 1998, 105 (10): 1849-1854. 10.1016/S0161-6420(98)91029-6.View ArticlePubMedGoogle Scholar
- Lim S, Wickremasinghe S, Cordeiro F, Bunce C, Khaw P: Accuracy of intraocular pressure measurements in New Zealand white rabbits. Invest Ophthalmol Vis Sci. 2005, 46 (7): 2419-2423. 10.1167/iovs.04-0610.View ArticlePubMedGoogle Scholar
- Mermoud A, Baerveldt G, Minckler DS, Lee MB, Rao NA: Measurement of rabbit intraocular pressure with the Tono-Pen. Ophthalmologica. 1995, 209 (5): 275-277.View ArticlePubMedGoogle Scholar
- Pallikaris Ioannis, Kymionis George, Ginis Harilaos, Kounis George, Tsilimbaris Miltiadis: Ocular Rigidity in Living Human Eyes. Invest Ophthalmol Vis Sci. 2005, 46 (2): 409-414. 10.1167/iovs.04-0162.View ArticlePubMedGoogle Scholar
- Cameron BD, Saffra NA, Strominger MB: Laser in situ keratomileusis-induced optic neuropathy. Ophthalmology. 2001, 108 (4): 660-665. 10.1016/S0161-6420(00)00577-7.View ArticlePubMedGoogle Scholar
- Katbab A, Pooyan S, Salouti R, Reza H, Hosseini J: High IOP as a cause of sudden increased myopia after LASIK. J. Cataract Refract Surg. 2005, 31 (10): 2031-2032. 10.1016/j.jcrs.2005.11.012.View ArticleGoogle Scholar
- Ahmadieh H, Javadi MA: Cilioretinal artery occlusion following laser in situ keratomileusis. Retina. 2005, 25 (4): 533-537. 10.1097/00006982-200506000-00027.View ArticlePubMedGoogle Scholar
- Kasetsuwan N, Pangilinan R, Moreira L, DiMartino D, Shah S, Schallhorn S, McDonnell P: Real time intraocular pressure and lamellar corneal flap thickness in keratomileusis. Cornea. 2001, 20 (1): 41-44. 10.1097/00003226-200101000-00008.View ArticlePubMedGoogle Scholar
- BingLi : Effect of pneumatic suction ring placement on intraocular pressure in cats. Ophthalmic Res. 2001, 33: 271-275. 10.1159/000055680.View ArticleGoogle Scholar
- Bissen-Miyajima H, Suzuki S, Ohashi Y, Minami K: Experimental observation of intraocular pressure changes during microkeratome suctioning in laser in situ keratomileusis. J Cataract Refract Surg. 2005, 31 (3): 590-594. 10.1016/j.jcrs.2004.06.085.View ArticlePubMedGoogle Scholar
- Acosta AC, Espana AM, Nose I, Orozco M, Yamamoto H, Venkatraman AS, Fantes F, Parel J-M: Estimation of rabbit intraocular pressure with commonly used tonometers. Ophthalmic Surgery Lasers & Imaging. 2007, 38: 43-49.Google Scholar
- Stoiber J, Fernandez V, Lamar PD, Hitzl W, Fantes F, Parel J-M: Ex-vivo evaluation of Tono-Pen and pneumotonometry in cat eyes. Ophthalmic Research. 2006, 38: 13-18. 10.1159/000088492.View ArticlePubMedGoogle Scholar
- The pre-publication history for this paper can be accessed here:http://www.biomedcentral.com/1471-2415/8/14/prepub
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.